Application area: Normal care
Product group: Wall supply units (decorative)
Elective service rooms for “very important patients”
A nursing room picture – this is how you could name the furnishing concept behind the VIP 2000, because here the high-quality room aesthetics are consistently in the foreground.
When closed the media connections are concealed behind a tasteful picture, which you yourself can select. Only when the picture is moved to the side with one hand movement, does the individually equipped connection technology appear.
The technology enables maximum flexibility in terms of connection equipment and the installation of additional accessories. For instance, pivoted extension arms and support arm systems for infusions and devices can be easily installed, making additional infusion and device trolleys superfluous.
Another highlight is that cables can also be fed out through the openings at the base of the picture, even when the module is closed. This means that even with continuous monitoring and supply, the room loses none of its comfort. There is also extra storage space for books and smartphone in the shelf segment integrated on the side.
Nominal voltage: 230 V - 240 V / 50 Hz - continuous operation
Protection class: I
Protection type: IP 20
medical gas technology
Operating pressure of medical gas technology
Oxygen: 5 bar
Compressed air: 5 bar
Vacuum: - 0,8 bar
Media current feed: Rear
Décor body: FUNDER-MAX 0570 FH textile grey
Décor sliding door HPL-laminate (digital printing)
Optional patient lamp Front-mounted
Assembly, cleaning, maintenance and repair
The medical supply units and single-bed lighting fixtures can be installed and mounted on solid or plasterboard construction walls. Customer provided substructure is required for plasterboard walls. Project-specific substructures are used for ceiling-mounted supply units.
Our products can be cleaned with conventional cleaning agents and disinfected with alcohol-free disinfectants.
Surfaces are protected by high-quality electrostatic powder coating. Non-coated parts are made of anodized aluminium or chrome-nickel steel. All plastic parts are resistant to cleaning agents and disinfectants.
The system must be maintained for the first time after 5 years and then after every 2 years. A contractual service agreement can be concluded for the maintenance work if required. More information can be found in the respective operating instructions.