Application area: Normal care
Product group: Wall supply units (classic)
The solution for narrow patient rooms
Sometimes narrow room dimensions limit the options for the use of horizontally arranged supply systems. So this where IVV 1054 lends a hand with a change of perspective. Rotated by 90 degrees, this vertical supply unit provides optimum patient care in the smallest of spaces. Arranged one above the other, the tapping points for medical gases, high and low voltage current and data and communications technologies can be designed as single, double, triple or four channel versions depending on your requirements.
But there‘s still more. Apparatus and accessories such as infusion bottle holders or examination lights all have their place on the IVV 1054 thanks to the stainless steel support arm fixed on the side making bulky apparatus trolleys or infusion stands superfluous.
As a result you not only gain space but also provide optimum working conditions for doctors and nursing staff.
modulux pure enhances the system as a matching lighting solution which like the IVV 1054 can be provided in electrostatically powder-coated variants in all RAL colours
Nominal voltage: 230 V - 240 V / 50 Hz - continuous operation
Protection class: I
Protection type: IP 20
medical gas technology
Operating pressure of medical gas technology
Oxygen: 5 bar
Compressed air: 5 bar
Vacuum: - 0,8 bar
Media current feed: Rear or from top
Optional support tubes: Side-mounted, rigid or swivel
Number of media channels: One to three
Total additional load: max. 100 kg/m
Assembly, cleaning, maintenance and repair
The medical supply units and single-bed lighting fixtures can be installed and mounted on solid or plasterboard construction walls. Customer provided substructure is required for plasterboard walls. Project-specific substructures are used for ceiling-mounted supply units.
Our products can be cleaned with conventional cleaning agents and disinfected with alcohol-free disinfectants.
Surfaces are protected by high-quality electrostatic powder coating. Non-coated parts are made of anodized aluminium or chrome-nickel steel. All plastic parts are resistant to cleaning agents and disinfectants.
The system must be maintained for the first time after 5 years and then after every 2 years. A contractual service agreement can be concluded for the maintenance work if required. More information can be found in the respective operating instructions.