Application area: Intensive care
Product group: Media column
More free space through freestanding placement in the room
With our MS 1054 media column which can be installed freestanding in the room, we offer you a flexible system for all room situations in which wall-mounted supply systems are out of the question.
The core of the system is a compact media column which is firmly anchored to the floor and ceiling and in which all required media accesses such as medical gases, high voltage current, data and communications technologies are integrated.
The optionally available accessories make the MS 1054 a multi-purpose and flexible treatment station. Side-mounted stainless steel tubes incorporate infusion bottle holders, lighting elements and other aids. But storage consoles and drawer pull-outs can be installed at any height you want – ideal for every work situation and always exactly where they are required in the room.
Nominal voltage: 230 V - 240 V / 50 Hz - continuous operation
Protection class: I
Protection type: IP 20
Lighting technology LED downlight (optional)
Nominal voltage: 95 V – 260 V / 50 – 60 Hz
Protection class: II
Operating type: electronic converter (1-3,6 W)
Lamp output: 1 W
Protection type: IP 20
medical gas technology
Operating pressure of medical gas technology
Oxygen: 5 bar
Compressed air: 5 bar
Vacuum: - 0,8 bar
Laughing gas 5 bar
CO2: 5 bar
AGSS: 5 bar
Media current feed: From the top
Optional support tubes: Side-mounted, rigid or swivel
Total additional load: Max. 160 kg
Assembly, cleaning, maintenance and repair
The medical supply units and single-bed lighting fixtures can be installed and mounted on solid or plasterboard construction walls. Customer provided substructure is required for plasterboard walls. Project-specific substructures are used for ceiling-mounted supply units.
Our products can be cleaned with conventional cleaning agents and disinfected with alcohol-free disinfectants.
Surfaces are protected by high-quality electrostatic powder coating. Non-coated parts are made of anodized aluminium or chrome-nickel steel. All plastic parts are resistant to cleaning agents and disinfectants.
The system must be maintained for the first time after 5 years and then after every 2 years. A contractual service agreement can be concluded for the maintenance work if required. More information can be found in the respective operating instructions.